You are in our Global Site
What Are the Main Functions of Each Part of the Shore Hardness Tester Verification Device?
Shore hardness is divided into two types: Shore indentation hardness and Shore rebound hardness. The former is to place the sample to be tested at an appropriate position on the hardness tester table,...
Use of Digital Force Meters
Ⅰ. The product use of digital dynamometerThe digital force gauge is a high-precision small portable tension and pressure testing instrument. Widely used in high and low voltage electrical appliances,...
Application of Coating Thickness Gauge in Plastic Industry
All coating thickness meter gauges are required to calibrate the zero position before measurement, which can be calibrated on the zero calibration plate accompanying the instrument or on the uncoated ...
Mikrometry cap torque tester is an intelligent multi-functional measuring instrument that is designed for testing various kinds of bottle lid torque. Bottle cap torque tester is mainly used in testing and calibrating tighten force, parts torsional destructive testing, etc. The bolt tension tester is easy to operate, with high precision and complete functions. Cap torque meter is widely used in various kinds of electric, light industry, machinery manufacturing, research institutions, etc. As one of the leading bottle cap torque tester suppliers in China, Mikro Metry provides quality bottle torque testers deserving your trust.
The cap torque meter tester is a kind of torque meter gauge designed to measure the applied and removed torque of bottle caps which can effectively grip containers of various shapes and sizes, while the optional flat and adjustable jaws provide a unique profile. Bottle cap torque tester made by Mikrometry, one of the leading measuring instruments suppliers in China, has a sturdy aluminum design and simple intuitive operation including a series of complex productivity-enhancing functions so that it can be used in laboratories and production environments.
ECT Bottle Cap Torque Tester Specifcations
Clamping Range: Φ19.5~Φ190mm
Curve or digital display modes
Calibration certificate Included
Charger with US/EUR/UK Plugs selectable
USB or RS232 communication interface, easy to communicate with PLC
Accuracy: ± 0.3% Readings
Sampling Rate: up to 2KHz
± 0.3% Readings
240*160 dot matrix LCD(3.1 inch) with LED Backlight
120% of F.S.
Track, Peak, First Peak、Auto Peak、Auto First Peak、Double Peak
Digital / Curve
Data Sampling Rate
Up to 2KHz
Programmable high and low limits
3.6VDC 1600mAH Ni-Mh rechargeable batteries
Approx. 25 hours per charge
Universal USB/BM charger, Input:110~240VAC
<0.03% FS per °C
USB or RS-232
Free software for USB communication
RS-232, "Within" set point output， "Under or over" set point output
Temperature: -10 to 40°C
Relative Humidity: 20 to 80%
Temperature: -20 to +50°C
Relative Humidity: 5 to 90%
~ 2.5 kg
AC adapter/charger, USB cable
Mini-printer, RS-232 Cable, Small Diameter Jaw, EST-TT2 Torque Test Stand.
0.1~2 / 0.0002
10~200 / 0.02
1~20 / 0.002
0.07~1.48 / 0.0002
0.25~5 / 0.0005
25~500 / 0.05
2.5~50 / 0.005
0.18~3.7 / 0.0005
2.2~44.3 / 0.005
0.5~10 / 0.001
50~1000 / 0.1
5~100 / 0.01
0.37~7.4 / 0.001
4.4~88.5 / 0.01
1~20 / 0.002
100~2000 / 0.2
10~200 / 0.02
0.7~14.8 / 0.002
8.9~177 / 0.02
Keystroke indication of ECT Bottle Torque Tester
LCD Screen Indication of Handheld Bottle Cap Torque Tester
1. Test Mode
2. The analog bar and direction on the ECT bottle cap torque tester indicate the current position in whole capacity. When the bar enters the area enclosed by a dotted line, means overload.
3. Storage/Print Setting
4. Time and Power
5. Torque Value and Unit
6. Statistics: Saved five-point measurements value.
7. Analysis: Average Variance criterion.
First Peak/Double Peak: Drop ratio
Mikro Metry handheld bottle cap torque tester can be used just put on or be fixed on the workbench or sand. For high capacity tests or accurate and stable measurements are required, this handheld bottle cap torque tester is best to use some kind of test stand or loader.
2. Power On/Off
Press the power button for 2 seconds to power On or Off the digital bottle cap torque tester.
3. Clamp Adjustment
Adjust the position of 4 clamping rods to fit the size of the test piece. This handheld bottle cap torque tester is best to turn on the bottle torque tester while adjusting and clamping.The overload alarm of this digital cap torque tester can be used to avoid overloading during operation.
① Select Units
② Select Test Mode
Select the appropriate test mode according to the measurement method or requirements.
③ Install test sample
④ Zeroing (remove tare)or Update(refresh measuring) Cap Torque Tester
Measured results can be stored in the handheld bottle cap torque tester so that you can review or print them later.
Other Torque Tester
The most accurate measurement results