How to Use Paint Thickness Gauge
Ⅰ. What is a paint thickness gaugeAs far as the construction industry is concerned, the thickness of the paint sprayed on the surface of the material is a very critical factor. So what tool is used t...
Factors That Cause Inaccurate Measurement of Coating Thickness Gauge
1. The strong magnetic field interferes with the measurement results of the coating thickness meterExperiments have proved that when the coating thickness gauge is working near an electromagnetic fiel...
Characteristics and Categories of Force Metre
Ⅰ. Features of push pull force gauge:1. Beautiful appearance, small size, easy to carry, the dynamometer has high precision and high resolution.2. Digital display, no parallax. Four measurement units...
EHS1A/5D is a high-precision shore hardness meter. This EHS digital shore hardness tester adopts high-precision displacement sensor and spring design with high test accuracy.
Shore durmeter meter/tester has three measurement modes: maximum mode, maximum average mode, and timing mode; 500 measurement data can be stored as one type of hardness test instrument, and the measurement data can be connected to PC software to export, built-in wireless printing function, you can connect wireless printer to print measurement data.
The digital shore hardness tester/durometer gauge price from our measuring instruments manufacturing company is very reasonable, welcome to order our durometer rubber roller hardness tester units. The shore A durometer gauge is used to measure soft plastic, rubber, synthetic rubber, felt, and leather. The shore A hardness test method is either hand-held for measurement or used in conjunction with the supporting load frame, and durometer hardness unit is the "degree". Different from the leeb portable hardness tester which is mainly used to test the hardness of metal materials, the shore durometer meter is focused on rubber and soft plastic hardness test.
Calibration certificate included
Conforms to ASTM D 2240, GB/T 531.1, ISO7619-1
A large, easy to read LCD display with a backlight
Measurement mode: Averaging mode, Max mode, Max-Average mode
Measurement data can be stored, browsing, and batch printing.
Printing: Wireless mini printer
Shore hardness testers are usually classified into following types. Digital durometer shore A is used for measurements below 90 degrees, and shore D hardness durometer tester is used for measurements above HA 90 degrees. Shore O hardness shore o durometer tester for rubber is used when the measuring degree is less than 20 degrees and the thickness of the measuring object is less than 3mm (the two are actually used less).
128 * 64 dot matrix LCD with backlight
1.5V *3 (AAA alkaline batteries)
-10 to 40℃，20 to 80% RH
150 * 50.5 * 29 mm
|EHS1A||A||Soft vulcanized rubber, natural rubber, nitriles thermoplastic elastomers. flexible polyacrylics and thermosets. wax, felt, and leathers|
|EHS5C||C||Medium-hard rubber, thermoplastic elastomers, medium-hard plastics, and thermoplastics|
|EHS5D||D||Hard rubber. thermoplastic elastomers. harder plastics, and thermoplastics|
|EHS1B||B||Moderately hard rubber, thermoplastic elastomers, paper products, and fibrous materials|
|EHSoo||OO||Soft rubber, thermoplastic elastomers, very soft plastics, and thermoplastics, medium-density textile windings|
|EHSo||O||Extremely soft rubber, thermoplastic elastomers. sponge extremely soft plastics and thermoplastics, foams, low-density textile windings, human and animal tissue|
|EHSdo||DO||Moderately hard rubber, thermoplastic elastomers, and very dense textile windings|
Other Hardness Tester
The most accurate measurement results