You are in our Global Site
What Are the Main Functions of Each Part of the Shore Hardness Tester Verification Device?
Shore hardness is divided into two types: Shore indentation hardness and Shore rebound hardness. The former is to place the sample to be tested at an appropriate position on the hardness tester table,...
Use of Digital Force Meters
Ⅰ. The product use of digital dynamometerThe digital force gauge is a high-precision small portable tension and pressure testing instrument. Widely used in high and low voltage electrical appliances,...
Application of Coating Thickness Gauge in Plastic Industry
All coating thickness meter gauges are required to calibrate the zero position before measurement, which can be calibrated on the zero calibration plate accompanying the instrument or on the uncoated ...
Desktop, small size, large torque
Dual sensors, wide measuring range
Sensor rotation measurement
Rotation angle display
7'' touch screen
|Measuring range||Sensor 1: 1～20 Nm||Sensor 1: 2.5～50 Nm|
|Sensor 2: 10～200 Nm||Sensor 2: 25～500 Nm|
|Resolution||Sensor 1: 0.002||Sensor 1: 0.005|
|Sensor 2: 0.02||Sensor 2: 0.05|
|Rotation angle||Sensor 1 : 45°00';Sensor 2 : 245°00'|
|The effective length of torque wrench||830mm (Max)|
|Display||7" touch screen|
Other Torque Tester
The most accurate measurement results